2mA High Power PORX Digital Portable Dental X Ray Equipment
Technical Data Rechargeable Lithium Polymer battery: 24V Maximum battery charge: 25.2V Battery current: 2mA, 60KV Battery Output: 10A Power supply: JBL7451250750100FJ, Bridge Power Corp. Input: AC100-240V, 50-60HZ 1.0A Output: DC25.2V, 0.75A Cable length of DC output: 1550mm Cable length of AC input: 1550mm Cable length of remote controller: 3000mm
X-Ray control and Generator Exposure time range: 0.01-1.6sec Maximum duty cycle: 1:60 Maximum inherent filtration: 1.6mm Al Maximum output power: 120W nominal at 60KV, 2mA Generator rating: 2mA at 60KV±5% Leakage technique factors: 60KV, 2mA, 1.6sec
Collimator cone Minimum source to skin distance: 10cm, 20cm (long cone) X-ray field size and configuration: 6cm diameter circle Shielding: led-line
Main use The unit is particularly provided for synthesis hospitals,cavity specialty hospital and dental clinic to take photo.
Environmental and working conditions
a. Condition of transportation and store Environmental temperature: 10~40°C Relative humidity: 5%-90% RH
b. Conditions of operation Environmental temperature: 10~30°C Relative humidity: 10%-85% RH
Classification/Specification Compliance MDD(93/42/EEC): AnnexIX [Class II b] IPX specification [IPO: do not operate under wet conditions] Type of protection against electric shock [Internal powered device] Degree of protection against electric shock [Type B applied part] For use in environments where no flammable anesthetics and/or flammable. Cleaning agents are present;non-alcohol based disinfectant only-wipes or cloth. Dampened with liquid/spray.