Korean High Frequency Portable Dental X Ray Machine
Type B applied part
X-ray generator *Tube voltage: 60kV (Fixed) *Tube current: 2mA (Fixed) *High voltage generating circuit: High frequency Inverter method *X-ray control method: Controlled by the micro processor *Time setting range: 0.01~1.60 second (0.01 sec step) X-ray tube *Type: Stationary anode X-ray tube *X-ray tube model: D-081BS(Toshiba) *X-ray tube focal size: 0.8mm *Target angle: 20° *Total filtration: 1.6mm Al (Inherent filtration: 0.8mmAl, Fixed added filtration: 0.8mmAl) Display method: LCD Display SSD: 100/200mm X-ray field: 55mm (Round style) Weight: 1.8kg Size of main body: 135(W)×150(D)×175(H)mm Voltage to use *Battery: Lithium polymer battery (DC24V) *Charger: JBL7451250750100FJ (Input: 100-240V~, 50~60Hz, 1.0A, Output: DC25.2V, 0.75A)
Instruction 1. Power On/Off switch: 2. X-ray tube current and voltage display: 3. Device operating condition or status display: READY, EXPOSURE, WAIT and ERROR 4. Indicates remaining battery charge: X-ray cannot be generated while charging or displaying battery information 5. Adult and child display 6. Anatomical teeth icon display: Incisor, canine and molar teeth of the upper and lower jaw (Maxilla and Mandible) 7. Exposure mode display: Film, sensor, and manual mode 8. Exposure time display: From 0.01sec. up to 1.60sec., 0.01sec. step time set. 9. Up and down control buttons: time set change (0.01sec. step) 10. Mode and Select button: Use to change and select the exposure mode.