Newest Design LCD X Ray Unit Dental Equipment Price
X-ray Rated Parameters
Tube Voltage: 70kV(fixed)
Tube Current: 2mA(fixed)
Time Set-up function: 0.05 ~ 1.20 sec
High voltage generation line: High frequency switch mode
X-ray control method: Microprocessor control
Type: Anodic fixed X-ray tube
X-ray tube focus size: 0.3mm
ANODE heating unit: 8.5KHu
Maximum use of tube voltage: 75kv
Target angle: 21°
X-ray exposure range: 73mm(Round)
Working voltage: Battery DC16.8V
Charger: AC100-240V, DC16.8V
Input Voltage: 100-240VAC
Output Voltage: 16.8 VDC
Do not repeat the camera in three minutes.
Do not use long-term, to turn off the total power switch.
Connect the battery DC socket to the DC IN socket on the back of the machine and confirm that the charger charging display LED is red.
Battery Charger Please use the machine-specific charger.
Batteries are consumables, and the use of time will gradually decrease over time with the initial purchase. For safety reasons, it is recommended to replace the battery for 2 years.
The battery can be used to charge more than 200 times.
When the power display is less than 14.8v, it is recommended to charge.
Each tooth for the shooting time reference: A. Teeth, the next anterior teeth 0.4, the anterior teeth 0.6, the next posterior teeth 0.8, the posterior teeth 1.0; B. Use the sensor in a time based on the reduction of 0.1-0.2 seconds.